A vector network analyzer (VNA) is an instrument that can be used to measure antenna or coax parameters such as SWR, impedance and loss. It can also be used to characterize and tune filters. It is a very useful tool to have if you are building and tuning home made antennas, filters or other RF circuits. For example if you are building a QFH or ADS-B antenna to use with an RTL-SDR, a VNA can help ensure that your antenna is properly tuned to the correct frequency. Compared to a standard SWR or network analyzer a VNA supplies you with phase information as well.
Until just recently, VNA's have cost roughly US$500 for a decent USB PC based unit like the miniVNA or PocketVNA, and have set people back thousands to tens of thousands of dollars for bench top units.
However, the cost of owning a VNA has now been reduced to only US$50 thanks to the NanoVNA. The open source NanoVNA project by @edy555 and ttrftech has been around since 2016, but only recently have Chinese sellers begun mass producing the unit and selling them on sites like Aliexpress, eBay and now Amazon. We note that it seems that there are some sellers selling them without shielding, so it might be worth double checking the listing to see if they mention that. All the listings we've seen seem to come with simple calibration kits as well.
The NanoVNA is a small credit card sized VNA. It has a built in LCD screen that can be used to display graphs directly, or it can also be connected to a PC and the graphs viewed via the NanoVNA Windows software. When purchasing you can opt to include a small battery for portable operation for a few dollars extra. The frequency range is from 50 kHz to 900 MHz, although you should note that above 300 MHz dynamic range performance is reduced.
Over on YouTube several hams and radio enthusiasts have recently uploaded videos demonstrating and reviewing the NanoVNA. The overall consensus is that the unit is accurate and works well. For additional support there is forum available at Groups.IO.
Below YouTube user IMSAI Guy reviews the NanoVNA. Check out IMSAI Guy's other videos too as he has several where he tests the NanoVNA on difference filters and antennas, and checks the accuracy.
Below is YouTube user joe smith's review. He also has two other NanoVNA videos on his channel where he shows how to use the NanoVNA to measure antenna impedance, and how to use the NanoVNA to create SPICE models for simulation.
Finally YouTube user Oli gives another overview. Please note that the following video is in Polish, but YouTube captions can be set to English.
We've also seen several recent text reviews:
NanoVNA - A short review. In this review nuclearrambo shows off the calibration kit, and shows a practical measurements of a directional coupler and 137 MHz QFH antenna.
NanoVNA compared with a Keysight fieldfox N9952A. Here nuclearrambo provides a comparison between the $50 NanoVNA and the $40,000+ Keysight FieldFox N9962A.
The NanoVNA, a real VNA at less than 48 €!. A review written in French, but Google Translate can be used. In this review David Alloza compares the NanoVNA against an Agilent E5062A benchtop VNA, and results look comparable.